Please leave this leaflet More information. Features PCA Patient Controlled Analgesia feature Patient Activated Timed and Continuous Infusion modes Clinician Override Bolus Volume ml and Mass Units infusion programming Accepts and automatically senses the size of syringes from the major manufactures Pneumatic patient handset dose cord – not offered on this auction Dose limit alarm Lockable security cover – this item do not need key. Mouse over to zoom – Click to enlarge. If the pump develops a fault then it must be referred to a suitably qualified engineer or returned to Smiths Medical in order to have the fault rectified. If the pumps internal AC supply voltage setting is altered both the label on the underside of the pump must be changed to indicate the correct AC voltage and the correctly rated fuse must be fitted.
|Date Added:||13 September 2015|
|File Size:||64.88 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
” + heading + “
If they fail to charge, all three batteries must be replaced at the same time. All other trademarks are acknowledged as the property of their respective owners Service Manual Issue 7 August Page i. The pump must not be immersed in any liquids or exposed to strong organic solvents. Pins 68 and 67 are held at Vcc and ground respectively as the reference inputs to the processor s built in A to D converter.
For many items we sell, we can arrange a demonstration either from our own team or the manufacturers. The contains four separate circuit boards: Taking the casing apart Each time the casing has been separated and reassembled, the syringe size functional tests detailed on page must be carried out.
The vast majority of pumps offered by sellers come with missing keys. The clutch, slotted disc and 3300 are located at the left hand end of the leadscrew, underneath the opto sensor board.
Resume making your offer if the page doesn’t update immediately. Remove and retain the two stringe and the bracket that holds the clutch and disc assembly in place. Specification Graseby pumps are subject to continual development and may, differ in detail from the following specification.
If the appropriate pulses are not detected by the opto-sensor then an alarm signal is generated. There are 2 items available. Getting your equipment to you swiftly and safely is of utmost importance to us.
MarCal Medical, Inc. – Improving the Way You Administer Medical Care
Learn syringr – opens in a new window or tab Postage: The clutch is held engaged by the spring with a pressure sufficient to overcome any resistance from the syringe plunger and the normal back pressure from the infusion line. The SETR is identical except for the color, wherein it has black switches on a black More information. Programmable 9 x 32 digit phone numbers for each trigger. Dartford Plug in slots K More information. Page x Issue 7 August Service Manual.
Industry leading returns on your unwanted devices. Learn more about your rights as a buyer. Starting by lifting a corner, peal the faulty panel away from the case. Push the connector and flexible lead of the new panel through the slot in the case. Purchase Peace of Mind. Be the first to review this product. It is safe, easy to use graeby effective for all types of PCA applications.
The pump must be set to display syrineg brand and size of syringe that is going to be used. IC17 provides a temperature compensated contrast control. During the fitting of the new leadscrew the spring assembly may have to be compressed to the left in order to fit it into the grooves in the rear casing. Remove the leadscrew assembly see page The alternative settings are as follows: If necessary, stick a new adhesive voltage rating label over the existing label on the underside of the pump.
Remove and retain the Spirol pin which will now be accessible.